Connectors suitable for a TRL calibration (microstrip)
I followed the MW101 encyclopedia page on TRL calibration. It strongly recommends 2.92mm air dielectric connectors. I have 2 questions.
1. When I search in the internet, I found some
2.92 mm connectors in the price range around US$ 50/connector. But the dielectric used is NOT air. How about using these for a TRL calibration?
2. All literature points out that 2.92mm (K) connectors are mechanically compatible with SMA and 3.5mm types. But in the internet, I found there are 2.92mm to SMA/3.5mm adapters. If these connector types are compatible, why do we need adapters?
Thanks a lot if you can clarify these.
The connectors are mechanically compatible, but you should not use a 3.5 mm cal kit on a 2.92 mm connector as you will have an electrical discontinuity at the reference plane. Rather you use the adapter to get the proper interface.
No, you should not use a dielectric filled 2.92 mm on a 2.92 mm cal kit for the same reason. Sounds like a bastardized 2.92 mm.
Now SMA cal kits are rare (as they are an abomination) so they expect you to use 3.5 mm on SMA. Though never connect an SMA to a 3.5 mm cal kit, especially an HP. Use a 3.5 mm jack saver.
It does seem weird that there are 2.92 to 3.5mm adapters available... how accurate do you need a calibration? if you don't work for NIST maybe you can ignore a slight discontinuity.
I would like to see the data sheet on the 2.92 mm connector with dielectric.... Thanks!
Keep in touch!
Sign in or register to add a comment.
Add a Comment
About Our Site
Biological Effects and Applications
Computer Aided Design
Emerging Applications and Technology
Filters and Passives
MMIC and RFIC
Packaging and Materials
Sources and Receivers
Test and Measurement
If 1.85 mm and 2.4mm connectors mate together then why would someone buy a 1.85mm to 2.4mm adaptor?
Why we use Multi-Line TRL calibration?
Seeking the original reference for a microstrip line impedance formula
Online Calculator to design Circularly Polarized Microstrip Antennas
Terms of Service
Useful Hints and Tips
Created with PlushForums
© 2023 Microwaves 101 Discussion Board