Short and Open Standard in Calibration Kit
Unicorn created the topic: Short and Open Standard in Calibration Kit
The textbook told me that it's hard to get open and short circuit over a broadband at microwave frequency, which is one of the reason why we use s-parameters instead of z-, y-, h-parameters(they all need short circuit to measure). But in the SOLT calibration of Network Analyzer, we do have short and open termination which can cover large frequency range. Is there any contradictory?
madengr replied the topic: Short and Open Standard in Calibration Kit
No. The SOLT coaxial shorts and opens are offset (i.e. moved from the reference plane) with a delay, and also have a polynomial fit for L and C, respectively. So they are far from ideal, but they are well characterized, thus the VNA can compensate.
MMWENGR replied the topic: Short and Open Standard in Calibration Kit
The only clarification I would add is that the VNA knows the polynomial fit coefficients since the cal kit and VNA are a matched pair. If the wrong cal kit is used to calibrate the VNA the accuracy is degraded.
Unicorn replied the topic: Short and Open Standard in Calibration Kit
Thank you for your comment.
It's not so much that perfect open or shorts are hard to create (though they are - but you can define that problem away), but many active devices are not stable with impedances far from 50Ohm - that is a physical problem rather than a mathematical transform...
Sign in or register to add a comment.
Add a Comment
About Our Site
Biological Effects and Applications
Computer Aided Design
Emerging Applications and Technology
Filters and Passives
MMIC and RFIC
Packaging and Materials
Sources and Receivers
Test and Measurement
Connectors suitable for a TRL calibration (microstrip)
Why we use Multi-Line TRL calibration?
Free Space Calibration Procedure
Terms of Service
Useful Hints and Tips
Created with PlushForums
© 2023 Microwaves 101 Discussion Board