Steps for connecting Femal-to-Female adaptor into a male connector in the RF measurement equipment
First of all, I am a newbie to the RF-mm-wave test equipments. I am trying to use this Agilent equipment shown in the image below where I want to connect a 1.85mm (F-to-F) sacrificial connector to the equipment's 2.4mm male connector. Could someone please guide me on how I should go about safely/properly make this connections? i.e. What steps I should take. For example, what part I should turn and what part I should be holding with the wrench
Also, once I make the connection, please also let me know on steps to disconnect this connection.
Start with tightening it by hand. You turn the large nut on the equipment (it’s a 2.4 mm NMD connector) while keeping the 1.85 mm adapter stable. Once it is finger tight, hold the 1.86 mm with a wrench and use a torque wrench on the NMD.
Sign in or register to add a comment.
Add a Comment
About Our Site
Biological Effects and Applications
Computer Aided Design
Emerging Applications and Technology
Filters and Passives
MMIC and RFIC
Packaging and Materials
Sources and Receivers
Test and Measurement
PA Input Protection - Pass Pulsed RF and block CW
About the additional metal plate for RF PCB (Ku band)
Any article/paper on use of high frequency signals ( up to 100Mhz) on General purpose connector?
If 1.85 mm and 2.4mm connectors mate together then why would someone buy a 1.85mm to 2.4mm adaptor?
Terms of Service
Useful Hints and Tips
Created with PlushForums
© 2022 Microwaves 101 Discussion Board